An oxygen concentrator is a medical device that delivers pure oxygen into the body of a patient by means of a nasal cannula or mask.
It takes in air from the surroundings, compresses it, removes the nitrogen from it via a filter and sieve beds leaving 87-95% of oxygen in the air.
This oxygen is then delivered to the patient.
Portable oxygen concentrator is an oxygen therapy device that can easily be carried around.
It offers therapy to those who are on the go and need more oxygen concentration than the ambient air level.
It is small in size and lightweight. Portable oxygen concentrators can even be carried on-flight by air travellers.
Stationary oxygen concentrators are larger and heavier compared to portable oxygen concentrators.
They also provide high flow rate and are recommended for patients who require substantial oxygen supply.
Stationary oxygen concentrators deliver 3 to 10 liters of oxygen per minute while portable models supply 1 to 5 liters per minute.
There are two types of portable oxygen concentrators - pulse dose and continuous flow.
Pulse dose oxygen concentrator is ideal for use in the day time and delivers air through a nasal cannula only when the user inhales.
This type is compact and has a greater battery life.
On the other hand, a continuous flow oxygen concentrator delivers constant air flow and is ideal for night-time use.
Pulse and continuous flow (or pulse dose and continuous flow dose) refer to how the oxygen is delivered to the patient.
A pulse flow machine delivers oxygen only when the patient inhales and is triggered by inhaling.
A continuous flow machine delivers a continuous flow of oxygen. Most continuous flow machines will deliver BOTH pulse and continuous flow.
This is a question that should be answered by your doctor. Most doctors recommend continuous flow for sleeping.
Yes, you can use paediatric flow meters that can reduce the litre per minute flow for paediatric patients.
Yes. Oxygen Concentrators that offer continuous flow can be used with sleep apnea devices.
Up to 94% - 95%.
The device is as simple to manage as it looks.
All that is required is that the coarse filter has to be washed each day and replace
the other filters as per instructions, which is usually at 1500 hours of use.
The filters may also need to be changed according to local conditions (dust, humidity) and the amount of use of the concentrator.
Skilled mechanical maintenance is needed only yearly.
There is a fourth filter, the bacterial filter, located before the oxygen outlet of the concentrator. Typically this must be changed every year, or more frequently, depending on the amount of use of the concentrator.
The concentrator may work 24 hours a day, every day, with a rest of 30 mins after every 8 hours of continuous use, as long as daily maintenance procedures are carried out. Every concentrator has an hour meter.
The working hours should be written down in the service report.
The humidity bottle must be cleaned every 2-3 days and distilled water should be filled to medium level.
Nasal cannula should be changed every three months for hygiene purposes.
The oxygen concentration is measured by an oxygen analyser which is available with all OXYMED service technicians. As the machine is used and the number of running hours increase and the machine comes to the end of its service life of 18k to 20k hours, the purity will also decrease.
The machine is fitted with Purity Sensor and will sound an alarm if it goes below 83%.
Purity can also be greatly affected if the machine is not maintained properly or if the suggestions in the manual are not followed.
There are mainly 3 sources of Oxygen for therapy:
The initial cost of the concentrator is Rs 45,000-50,000, which may be considered expensive in comparison to cylinders.
Nevertheless, the yearly running cost of Rs. 12,000-15,000 is substantially less than the Rs 1.5 to 3 lakh for providing cylinders and the Rs 1.2-1.3 lakhs for liquid oxygen
Thus on cost alone the oxygen concentrator is the preferred means of providing oxygen treatment.
All OXYMED machine are fitted with Oxygen sensors which are calibrated to sound an alarm if the oxygen purity goes below 83%.
If such a problem occurs, please see the manual for troubleshooting.
The life of the zeolite crystals (Sieve Beds) are the heart of the machines and can be expected to be at least 20,000 hours which in most situations would give about 3-5 years use.
This life of 20,000 hours depends on the running environment of machine, timely changing of filters and general regular maintenance and upkeep of the machine.
Patient must follow the precautions and suggestion given in the manual for long life of the concentrator.
Most people prefer nasal cannula as it doesn’t look medical but most cannulas have a limitation that they are effective for flows up to 6 Litres Per Minute (LPM).
There are special cannulas for higher flow (you may check with your equipment provider).
However, if you get rashes in your nostrils by use of cannula or if your flow requirement is above 6 LPM then a simple face mask is used by most people.
It is recommended to use clean drinking water (distilled water) for humidifier as after all you are going to breathe through it.
There is no hard and fast rule for cleaning but there is no harm in having good daily cleaning practices just by rinsing through fresh water and replacing the water in the bottle.
The purpose of humidification is to provide moisture which makes Oxygen therapy comfortable; if not used, can cause dryness in airways of nose and throat. Usually, use of humidifier is recommended in continuous flow of more than 3 LPM
For patient comfort, please use the humidifier bottle with water up to mid level of bottle.
You feel this because the temperature of Oxygen coming out of the machine is exactly the same as room temperature.
However, this is proven that effectiveness of Oxygen from both the sources is virtually the same.
What you need to see is the Oxygen saturation in your blood and you can check this using an Oximeter.
Please consult your physician immediately.
The ideal amount of Oxygen is decided by your physician; he does it by carefully analysing your ABGs.
In most cases, the goal is to keep your Oxygen levels above 88% which after prescription is monitored using Oximetry.
In healthy individuals, ideal saturation of Oxygen in blood is 95-99%.
No, you should use Oxygen exactly as prescribed by the physician.
Using less than prescribed Oxygen can cause low levels of Oxygen in blood which can further cause poor memory, irritable mood, poor cognition, shortness of breath, headaches and much more.
Using less than prescribed oxygen for longer periods of time also deteriorates lung's health and causes the disease to grow.
No, using more than prescribed oxygen can cause Oxygen toxicity/Oxygen poisoning in the body which has many harmful effects like damage to eyes, acute respiratory distress syndrome (ARDS), alveolar damage, destruction of neurons , etc.
No, Oxygen is not addictive at all. A large number of patients express their concern regarding this and try to decrease the time they are on Oxygen or totally avoid it.
Supplemental Oxygen is prescribed because the lungs have become weak due to condition like COPD and the 21% Oxygen of air is now not sufficient for lungs which results in low level of O2 in blood.
In order to make lungs feel like as if they are exactly in a normal healthy environment, they need some extra Oxygen so that the Oxygen exchange with blood is exactly same as it would have been with normal healthy lungs.
Who do I call if I have a question regarding how my unit is working?
First, you can visit our website and browse through the help section for your questions.
Second you can download our MEDEQUIP APP from the app store and get access to the database where most of the questions can be answered.
Thirdly you can call our service technicians PAN India with the contact information given below.
" Always contact authorized service centre for service / Repair. Oxy Med machines are backed by original Spares replacement, Where none of spares are repaired but only replaced with new "
Start by identifying your Warranty Card and Invoice of the concentrator.
These are most important to claim the warranty.
Once you have these two documents, please proceed to either register on MEDEQUIP APP or by calling the nearest service station.